Accuracy of Bioelectrical Impedance Consumer Devices for Measurement of Body Composition in Comparison to Whole Body Magnetic Resonance Imaging and Dual X-Ray AbsorptiometryBosy-Westphal A. · Later W. · Hitze B. · Sato T. · Kossel E. · Glüer C.-C. · Heller M. · Müller M.J.
a Institut für Humanernährung und Lebensmittelkunde, Christian-Albrechts-Universität, Kiel, Germany bOmron Healthcare Co., Ltd, Kyoto, Japan c Klinik für Diagnostische Radiologie, Universitätsklinikum Schleswig-Holstein, Campus Kiel,Germany
Objective: To compare body composition determined by bioelectrical impedance (BIA) consumer devices against criterion estimates determined by whole body magnetic resonance imaging (MRI) and dual energy X-ray absorptiometry (DXA) in healthy normal weight, overweight and obese adults. Methods: In 106 adults (54 females, 52 males, age 54.2 ± 16.1 years, BMI 25.8 ± 4.4 kg/m2) fat mass (FM), skeletal muscle mass (SM), total body bone-free lean mass (TBBLM), and level of visceral fat mass (VF) were estimated by 3 single-frequency bipedal (foot-to-foot) and one tretrapolar BIA device, and compared to body composition measured by MRI and DXA. Bland-Altman and simple linear regression analyses were used to determine agreement between methods. Results: %FMDXA, SMMRI or TBBLMDXA showed good relative and absolute agreement with two bipolar and one tetrapolar instrument (r2 = 0.92–0.96; all p < 0.001; mean bias <1.5 %FM and <1 kg SM or TBBLM) and less relative and absolute agreement for another bipolar device (r2 = 0.82 and 0.84, mean bias ∼3 %FM and ∼3 kg SM). The 95% limits of agreement (bias ± 2 SD) were narrowest for the tetrapolar device (–6.59 to 4.61 %FM and –4.62 to 4.74 kg SM) and widest for bipolar instruments (up to –14.54 to 8.58 %FM and –9.52 to 3.92 kg SM). Systematic biases for %FM were found for all bipedal devices, but not for the tetrapolar instrument. Conclusion: Because of the lower agreement between foot-to-foot BIA and DXA or MRI for the assessment of body composition in individuals, tetrapolar electrode arrangement should be preferred for individual or public use. Bipolar devices provide accurate results for field studies with group estimation.
Dr. Anja Bosy-Westphal, Institut für Humanernährung und Lebensmittelkunde, Christian-Albrechts-Universität zu Kiel, Düsternbrooker Weg 17, 24105 Kiel, Germany, email@example.com
Published online: December 2, 2008
Number of Print Pages : 6
Obesity Facts (The European Journal of Obesity)
Vol. 1, No. 6, Year 2008 (Cover Date: December 2008)
Journal Editor: Hebebrand J. (Essen)
ISSN: 1662-4025 (Print), eISSN: 1662-4033 (Online)
For additional information: http://www.karger.com/OFA