Automated Analysis of Data Is Inferior to Visual Analysis of Ambulatory Sleep Apnea MonitoringFietze I. · Glos M. · Röttig J. · Witt C.
Department of Cardiology, Angiology and Pulmology, Humboldt University Medical School Charité (Campus Mitte), Berlin, Germany
Do you have an account?
- Rent for 48h to view
- Buy Cloud Access for unlimited viewing via different devices
- Synchronizing in the ReadCube Cloud
- Printing and saving restrictions apply
Rental: USD 8.50
Cloud: USD 20.00
Background: Many ambulatory sleep apnea monitoring devices are equipped with software which allows an automated analysis of data as well as a visual analysis. Objective: The Merlin system which records heart rate, snoring sound, efforts, oronasal flow, body position and oxygen saturation was investigated to identify proper parameter settings for the automated analysis and to compare the automated with the visual analysis in patients with mild obstructive sleep apnea syndrome (OSAS). Sensitivity and specificity of the visual and automated analysis of ambulatory monitoring in comparison with visual polysomnographic (PSG) analysis were determined. Methods and Results: First, we tried to find the optimal parameters for the automated analysis, using 7 different settings in 17 OSAS patients. Furthermore, we applied the optimized setting to 66 OSAS patients who were admitted (age 50.9 ± 9.9 years, BMI 32.9 ± 5 kg/m2), and compared the results with the visual analysis of raw data. The patients slept for one night in the sleep laboratory with Merlin and PSG simultaneously to compare the visual and automated analysis of Merlin data with results from the visual analysis of PSG. Automated analysis leads to an underestimation of the respiratory disturbance index (RDI; p < 0.001) compared with both the visual analysis and results of PSG. Using a cutoff level of 5 apneas and hypopneas/h for the diagnosis of OSAS, the sensitivity of Merlin with the automated analysis is 40.6% and the specificity is 100%. With a cutoff level of 15/h, sensitivity and specificity rose to 91.3 and 100%, respectively, which is comparable to the visual analysis. Conclusion: Merlin is a reliable device for detection of sleep-related breathing disorders, but recordings should be analyzed visually, especially in patients with a low RDI.
© 2002 S. Karger AG, Basel
Article / Publication Details
Copyright / Drug Dosage / DisclaimerCopyright: All rights reserved. No part of this publication may be translated into other languages, reproduced or utilized in any form or by any means, electronic or mechanical, including photocopying, recording, microcopying, or by any information storage and retrieval system, without permission in writing from the publisher.
Drug Dosage: The authors and the publisher have exerted every effort to ensure that drug selection and dosage set forth in this text are in accord with current recommendations and practice at the time of publication. However, in view of ongoing research, changes in government regulations, and the constant flow of information relating to drug therapy and drug reactions, the reader is urged to check the package insert for each drug for any changes in indications and dosage and for added warnings and precautions. This is particularly important when the recommended agent is a new and/or infrequently employed drug.
Disclaimer: The statements, opinions and data contained in this publication are solely those of the individual authors and contributors and not of the publishers and the editor(s). The appearance of advertisements or/and product references in the publication is not a warranty, endorsement, or approval of the products or services advertised or of their effectiveness, quality or safety. The publisher and the editor(s) disclaim responsibility for any injury to persons or property resulting from any ideas, methods, instructions or products referred to in the content or advertisements.